Thank you sir for the best answers.
I have another question to ask.
My new MPAES 4210 is just a year old. My plan is to Analyse metals for Lube Oil Condition Monitoring for mine fleets.
However, I find bit difficult to get the correct calibration. I used two-point calibration standards as 0ug/g, 100ug/g and 900ug/g (for Additives). MPAES software error message signal me as;
1. QC setup failed.
2. QC calibration validation failed
Hence, seeking any advice to get the MPAES calibration set up for Organic Lube Oils.
Thank you for your kind assistance