I'm trying to use the DA Reprocessor (B.08.00) to process TOF data as a batch on a Windows workstation. If there are too many features, the DA Reprocessor stops with the following error in the log: Failed to run worklist Qual: The Operation has time out. I know the processing will work if I run it in MassHunter Qualitative Analysis, but I… Show more
Choose an area for discussions and documents
Hi, I need process Residual Solvents, using software OpenLab Ezchrom Version A 04.07, but i don't know how retired results in software. Normally i used using Excell for calculete this. How to calculeted this in software? Equation 4(C/W)[Ru/(Rs-Ru)] C: concentration residual solvents in standard w: weithg sample Ru: area residual… Show more
Hi ALL, So I finally got my 1100 stack to communicate with Xcalibur and run a method and write a raw file in Xcalibur. How do acquire or view DAD spectral data during such method runs? Thanks, Billy
I suspect a firmware issue. When chemstation is attached it list FW as: Xcalibur "Getting Connected" states it requires firmware. A.06.01  Chemstation lists firmware for units as: Diode-array detector G1315A A.06.32  Quaternary pump G1311A A.06.04  Thermostatted… Show more
I have a 6890N and was doing S/SL inlet maintenance yesterday when the pressure "actual" for the S/SL changed to 152. This occurred after pressing "Off" on the S/SL pressure and then disconnecting the column from the inlet. The value was stuck at 152 and would not change. I rebooted the GC a few times to no avail. Finally after unplugging the GC… Show more
- mmehling likes james_jenkins's reply Re: Method Resolution for Changed Configuration?
- nyirendas likes rolanddiaz-bone's document Collection of Support Documents
- james_jenkins likes johnbradley's reply Re: Quantitation Frustration!
- james_jenkins likes m4esti's reply Re: 5973A MSD HPIB communication
- michalab likes ryoboyle's reply Re: Intelligent Reports
Load more items
The autosampler showed an error message "fatal error", and the machine stopped working with a pressure around 250bar, no flow. The machine worked normal if we restarted it. But later, it stopped again showing the same message. Did anyone have met the same problem? Thank you.