This Information Applies To: Agilent ICP-MS (7500/7700/8800)
How to replace the Forline Pump Oil in the E2M18 or E2M18F?
The E2M18 oil should be replaced at least every three months or when the oil color indicates it needs to be changed.
The E2M18F (Fomblin model) oil should be replaced every six months. Note, the Fomblin Oil will not change color even when it is old/dirty.
Check the foreline pump oil color (if applicable) and level monthly.
Steps to Follow
Learn how to effectively maintain your Agilent ICP-MS System:
Agilent 7800/7850/7900 ICP-MS with MassHunter and Agilent 8900 Triple Quadrupole ICP-MS with MassHunter e-learning courses available from Agilent education