This Information Applies To: Agilent High Performance (Well-Plate) Autosampler G1368A, G1377A, G1367A/B/C/D/E, G4226A, G5667A, Agilent Multisampler G7167A/B, G5668A, G7137A, G4767A, Agilent OpenLab CDS 2.x, Agilent OpenLab Chemstation C.01.xx.
This article outlines the procedure on how to set up custom wellplates in Agilent High Performance Autosampler or Multisampler via Agilent OpenLab software.
Steps to Follow
Agilent Chromatography Software contains a list of all available wellplate types, including standard manufactured plates and custom wellplate types. If the wellplate type you want to use is not included in the list, you can add it using the Define and Edit Wellplates dialog box to edit an existing wellplate definition or set up a new one.
|Learn how to effectively operate your Agilent samplers:
Multisamplers Agilent InfinityLab LC Series user manual on Agilent.com
1290 High Performance Autosampler InfinityLab LC Series user manual on Agilent.com
1260 Bio-inert High Performance Autosampler InfinityLab LC Series user manual on Agilent.com