Thanks for your help in advance!
Our G6230 TOF was suddenly vented by itself once the instrument detected "insufficient high vacuum ". So, today morning I restarted TOF instrument and performed the manual-recommended start-on (pumping down + 8 hr HV conditioning) using a diagnostic tool program.
Then, I found out that there is no vacuum reading from Turbo 1 at the parameter section from the diagnostics tool.
However, I can see TOF vacuum reading at the SC readbacks section.
Are there anyone who is familiar with this issue?
I also wonder whether this issue is related to the error showing up last Saturday or not.