This Information Applies To: Agilent 6400 Series triple quadrupole LC/MS systems equipped with a gate valve
Installing the Ion Transfer Capillary after cleaning or replacement in Agilent LC/MS instruments with a gate valve.
Clean powder-free nitrile gloves
This Information Applies To: Agilent 6400 Series triple quadrupole LC/MS systems equipped with a gate valve.
Removing the Ion Transfer Capillary for maintenance on instruments with gate valve. The gate valve allows ion capillary removal and installation without having to vent the instrument...
This Information Applies To: Agilent 6000 Series LC/MS systems equipped with MS40+ rotary vane pumps
Instructions on how to replace the filter in an Agilent MS 40+ pump as part of annual pump maintenance
The filter in the MS40+ pump should be replaced every 12 months of operation...
This Information Applies To: Agilent 6100 and 6400 Series LC/MS Systems
Tuning a mass spectrometer involves optimizing the voltages applied to various optical elements in the ion path to achieve the required mass accuracy, resolution, and sensitivity that the user requires for their chosen...
This Information Applies To: Agilent 6400 Triple Quadrupole LC/MS systems, MassHunter Workstation 10.1 or earlier. This article does not apply to MassHunter Workstation 12.0.
Users need to reconfigure a new LC/MS triple quadrupole after a new MassHunter software installation.