This Information Applies To: AriaMx real-time PCR system
Error 43 is displayed on the AriaMx touch screen. Error 43 is a voltage error and it normally indicates that the cartridge LED voltage or motor voltage being detected is abnormal. Sometimes it can be solved by rebooting the AriaMx instrument and reseating the optical cartridges, otherwise it would usually require a repair.
Steps to Follow
|Caution: Once you have removed the optical modules, be sure not to place them down with the filter windows on the bottom in contact with the bench to avoid the windows getting scratched. Therefore when you place the cartridge on the bench just lay it on its side.|
If the error persists, run system diagnostics and contact your local Agilent technical support with the exported diagnostic results in .res format. You need to be logged into the AriaMx software as Administrator to run the diagnostics and export the .res file.
|Learn how to effectively operate your Agilent AriaMx:
AriaMx Real-Time PCR System Setup and User Guide on Agilent.com
Aria Real-Time PCR Software User Manual on Agilent.com
For Research Use Only. Not for use in diagnostic procedures.