This Information Applies To: Agilent AriaMx real-time PCR system, AriaMx optical modules
AriaMx users can change which AriaMx optical modules are installed to upgrade the instrument capabilities or to troubleshoot.
Any AriaMx optical module:
Steps to Follow
Caution: Finger prints or scratches on the glass window of the optical module (the surface opposite of the label) may impact its functionality. Avoid touching the glass window of the optical modules. Do not put the optical module down with the glass window in contact with the lab bench or other surface.
The optical module configuration can be changed either with the AriaMx switched on or off.
AriaMx is Switched On
AriaMx is Switched Off.
Caution: Only move the optical module housing carrier by hand if the AriaMx is switched off and the power cable is disconnected
For Research Use Only. Not for use in diagnostic procedures.