I have a question.
Can I connect two-column to an ms detector? can i use a tee? will it be safe for the turbo pump because the flow will increase?
Hi azrael, as mhardacker has posted, a two hole ferrule works well; but they are more prone to leaks than a single hole ferrule, and can let in a lot of air on temperature cycling. Could I recommend the…
I think what possible. But need to consider several parameters: flow, id column, column temperature range....see also the topic change the column gc/ms without stopped turbo pump
the simplest solution is a TWO hole ferrule, whereby both columns must have the same outer diameter.
see for this: https://www.agilent.com/store/en_US/Prod-5062-3580/5062-3580
for the purpose…