During volume accuracy test of our 850-DS Dissolution Sampling, the difference of the sample volume for position #2 for 10mL was higher than 0.5mL, being that specification is lower than 0.25mL.
How can we calibrate the equipment to sample the right volume?
We performed the full range volume dispense (in the calibration screen), but we are not sure how to use the data to do the equipment calibration.
Could you help us, please?
|#||Empty Test Tube||Filled Test Tube||Difference||Actual Sample Volume||Difference from Set Point of 10 mL|
Thank you so much.