This Information Applies To: Agilent GC systems
Before most maintenance procedures, the GC must be set to safe temperatures, pressures, or flows. You can create and save the maintenance method in the controlling software or the GC itself. A proper maintenance method will protect the user...
This Information Applies To: Agilent 7683B Automatic Liquid Sampler
O nly use this alignment procedure if the injector hardware configuration has changed or if the Align Mode light comes on.
Agilent does not recommend performing this procedure unless the Align Mode light turns on, or if...
This Information Applies To: 5977A and 5977B
Content: This procedure is to guide the installation of a capillary column on 5977A/B GC/MS with a standard column nut or self-tightening nut.
Warning : a. The GC and MS operate at high temperatures. Do not touch any parts of GC and MS until...
This Information Applies To: Agilent 6850 A/B, 6890 A/N/Plus, 7820A, 7890A/B, 8860, 8890 Gas Chromatographs with Flame Ionization Detector (FID)
The cleaning of the Flame Ionization Detector (FID), may be required in the following scenarios:
Preventive or periodic maintenance.
This Information Applies To: All Agilent GC systems with, µECD detector
The micro Electron Capture Detector (µECD) shows a higher than usual baseline signal.
Warning : The ECD utilizes a radioactive source.
You may not open the Electron Capture Detector (ECD)...