We have a third party detector with a voltage signal output that we are currently using with an Agilent 1290 Infinity I system (ChemStation software). We are using a G1390B 1200 Infinity Universal Interface Box II (UIB), connected to one of the CAN ports on the 1290 pump, to receive the signal into Chemstation.
We would like to move this detector/UIB setup onto an Agilent 1260 Infinity II system, with Masshunter Acquisition 8.0 software. I would like to know if the Masshunter software would be compatible to add the UIB module? If so, can we expect the UIB to be auto-detected once we open the Instrument Configuration tool and click to configure the 1260, or are there any steps we will need to take to manually add it?"