I am using Agilent 1260, with Vialsampler G7129A. After every few injections now, after the Vialsampler injects the sample, the Vialsampler turns red and shows the error message as hardware error[EE 34331,0].
This has been a recurring issue with us and we cannot seem to identify the source of the problem.
As this is happening after every few injections, it is getting difficult to run sequences as it may stop at any injection.
If anyone has faced similar issues or could have any idea we could avoid this problem, please help us resolve this.