we perform column exchange, gas exchange and clean the ion source. therefore, after the equipment was restarted, an autotune was performed on December 15th (report attached), some parameters were high, mainly accuracy. The equipment was in conditioning until December 18th, where another autotune was made. The errors below were detected with the following report.
Our equipment is a GCxGC (Agilent 7890) with Q-TOF-MS/MS (Agilent 7250)
Can you help me?