i got baud rate error from 8000 diss. sampling station and syringe pump is not working. i added image of the error.
i really dont get what is wrong.
The baud rate should be set to 9600.
On the 8000, press PRINT.
Select #5 "Set Baud Rate".
Set the baud rate to 9600.
Below is a link to the user manual describing this in more detail: