This Information Applies To: Agilent MassHunter Qualitative Analysis Software, Revision 7 and up
This article shows the procedure to change the number of decimal places in the Mass Spectrum Preview and MS Spectrum Results windows of the MassHunter Qualitative Analysis. The changes in these windows will appear in the m/z column in the MS Peaks table.
Steps to follow
In the top menu, click Configuration > MS and MS/MS Spectra Display Options
Alternatively in the MS Spectrum Results window tool bar, click the Spectrum Display Options icon .Tips: Changes made in this window are also viewable in the MS Peak Lists.
Adjust the number of digits in the Digits after the Decimal field.
Figure 1. MS and MS/MS Spectra Display Option window
Click OK to save the change and to exit the dialog box.
To see the changes, review the MS Spectrum Results window.
Figure 2. MS Spectrum Results Windows showing two digits after the decimal
Learn how to effectively operate your MassHunter:
Agilent 5977 GC/MSD with MassHunter Workstation
Agilent 7000 or 7010 Triple Quadrupole GC/MS with MassHunter Workstation e-learning paths available from Agilent Education