This Information Applies To: Agilent OpenLab CDS
Outlining the step-by-step instruction on how to integrate peaks using Integration Events in Processing Method.
Integration locates the peaks in a signal and calculates their size.
There are two ways to integrate peaks in OpenLab CDS:
Data Analysis includes the same integration algorithms that are available in both OpenLab CDS ChemStation Edition and OpenLab CDS EZChrom Edition. The processing method supports all the initial and timed events from ChemStation or EZChrom. In each method, the preferred integrator must be selected.
To be able to carry out the procedure as described, the privilege of Processing Method > Edit general parameters is required. Privileges are configured in the Control Panel.
If the project enforces method approval, the method status must be Generic.
For details, see Method approval on the Agilent OpenLab Help and Learning page.
Steps to Follow
For more details, see user resources from OpenLab Help and Learning:
|For more details, see Configure signal-specific integration parameters on the Agilent OpenLab Help and Learning page.
For more details, see Graphically change integration events on the Agilent OpenLab Help and Learning page.
For more details, see Advanced Integration Events, from OpenLab Help and Learning.
Learn how to effectively operate your Agilent OpenLab CDS software:
HPLC-OLII-1140eV2 - Agilent Infinity LC systems with OpenLab CDS - Integration and viewing results (Ver. 2.6) e-learning course available from Agilent education