We have a 850-DS dissolution sampling station in our lab and it is not completing and the light on board is flashing when cleaning procedure (in the picture).
I aborted it on Dissolution Workstation Software to finish cleaning procedure. I restart system and tried again, but this problem is still happening.
Is there any part in the system that would be needed to be replaced??
Thank you so much/