Hello everyone, thank you for this support forum.
I am new to Agilent and Chemstation software and i need some tips.
I set the method and put values in calibration table.
After the sample rum the report shows me Ret Time, Area, Amt/area and Amount %. If I have understood correctly:
Area is the sample Area
Amt/area is the calibration factor from calibration table (mass_standard / Area_standard)
Amount % is Area * Amt/Area
The EN standard that i have to use indicates that in order to calculate the result I have to make this calculation:
Now the question is: where to insert *25? in multiplier?
And if I wanted the software to do the complete calculation where would I enter the sample mass? in the sample info before starting the analysis?
I tried with no result... please help!