Normally the weight of the standards is entered in the sequence table in the column of compound amount (amount in Sample). The problem is that often this weight must be corrected for purity and water content, these data are variable depending on the batch of standard used.
How could we perform the calculation so that the corrected weight is the one used in the calibration graph in the report?
We are using Openlab CDS 2.3.
Thank you very much!