Choose an area for discussions and documents
The sensitivity of GC-FID 7890A has low sensitivity.
Followed the LAN setup in the manual, yet all I get is Powerfail. Using Chemstation version B.05.04 (which is supposed to work on Windows 10).
Hi! Due a change to OpenLab CDS C.01.10 the setup of the sequence changes. For the old Version i setup a Sequence with a macro using the commands: SetTabVal _SEQUENCE, T$, R, "Vial", Vial Where T$ is the "SeqTable1" od "SeqTable2". So i can set up a list for front and back and then start the sequence. In Version C.01.10 the Sequence is… Show more
Hi all, We utilize various different methods and prefer to queue sequences up whenever multiple methods are needed for our analyses. Problem is, whenever the 1st sequence is complete, the instrument sits idle and does not start the next (queued-up) sequence. It appears that you have to click the save icon (floppy disk) to get it started,… Show more
- cmcdonal likes yli594's question Leaking from needle seat
- wydnaa likes elodie's reply Re: How to Learn OpenLab CDS ChemStation
- rolanddiaz-bone likes mnevill's document Collection of Cell Analysis Resources
- fpmlab likes gareth.pearson's reply Re: Help Fixing Low Sensitivity Issues
- james_jenkins likes gareth.pearson's reply Re: Cheminert Valve for Agilent 7900
Hi, I am currently using a 1260 ALS autosampler (6 port valve) on my HPLC system. However, every time I inject sample, or simply run my method at "mainpass" mode for the injection port, I can see a significant pressure drop and also leaking from the needle seat. I just changed a new needle seat, needle, rotor seal and isolation seal. But it is… Show more
Load more items
Hello, Has the compatibility of ChemStation B.05.02 been tested on Windows 10? Alternatively has it been tested in the Windows 10 compatible mode? I am trying to decide if the computer can be upgrade from Windows 7 to 10. Thank you for your help, Kang