Using GC-MS - Single Quad?
is there any difficulty encountered?
I typically do this was a pulsed splitless injection. 50 psi pulse for .5 min and purge flow to split vent of 50ml/min at .48 min. I typically use a single tapered direct connect liner in the GC inlet.
Retrieving data ...