Hi there, I'm in the process of replacing the source assembly for the FTIR Cary 630, but I cannot locate the part to purchase. Could you please direct me to the item I need to order? Thank you!
The part number of the part : Source with purge fitting G8043-67400
Remember, you need to purge the instrument after replacing the source.
Thank you so much for this! The purge is new...I do not see that step in the users guide after replacing the source - it just to reinstall and to check the energy values and adjust the gain if needed. Is there additional instructions for purging I need to read up on?
Retrieving data ...