I am a new in user if chemstation.
What version of ChemStation are you running? You can determine this by going to Help>About.
Version of Chemstation B.03.
İ tried but i didn't get it.
Since intelligent reporting is not available in B.03.xx revisions of ChemStation you will be limited to using the classic sequence summary report style. To setup for a sequence summary report to be generated for a given sequence, you will need to go to Sequence>Sequence Parameters>Sequence Output or possibly just Sequence>Sequence Output (I don't have a B.03.xx system to reference, but it will be one of those two options). In Sequence Output you will need to select the option to "Print sequence summary report" and then setup the report appropriately. Depending on what you want RSD calculated for you will need to select the option for statistics calib. runs and/or statistics sample runs in the setup menu. Standard statistics cover retention time, area, height, amount, peak width, and symmetry. If you wish for summary calculations to performed for other results then you will need to select Extended Statistics.
Once the sequence output is configured you will need to save the sequence. Then if this is for a sequence that has already been acquired you will just need to reprocess the result set to generate the summary report.
Thank you sir for your explanation.
Retrieving data ...