Hi Folks, currently we use factors instead of a calibration in EZChrom regarding our I.S. method, (see attached). How would we translate this into an Openlab 2 I.S. method?
I have added tags to this post to increase visibility.
From your attachment it looks like the STD Mult you are specifying for your ISTD compound is 1, meaning that in your method STD Mult is just another correction factor that is multiplied during the concentration calculation:
In OpenLab CDS you can accomplish the same thing by using the Multiplier column in the processing method compound table:
Thanks for that I'll take a look tomorrow and get back to you. We have just had openlab 2 installed and it's been a rush to get everything up and running.
Retrieving data ...