I am in the US and one of our plants in France will be shipping their ELSD (G7102A) to me for troubleshooting. I realize the regional voltages are different. What is the best way for me to power this module? Thanks in advance!
The manual mentions:
Thanks! I saw that and there is also a section about checking which fuse is in place. I don't know if I can change the fuse, then use a 110V line or not. I also have a 208V line in the lab, but I don't know if that will be compatible or not. I am sure even if it is, the plug will be different.
All you need to do is to replace the French power cable with a proper US one.
Excellent news! I was hoping it would be that easy, but wasn't sure. I can just use the cable for the one I will be taking out of service for the troubleshooting. I have MANY proper Agilent US power cords! Glad I don't need to do anything with a fuse. THANK YOU!
Retrieving data ...