A colleague in our QC group has a newly purchased Agilent GC unit. Prior units used a SATIN interface to link the GC unit to their EMPOWER software system. The SATIN box converts the picoAmps signal to microVolts (which is what EMPOWER uses to measure peak area).
My colleague is looking for an Agilent point of contact to discuss potential solutions to this issue.
(PERSONAL DETAILS REMOVED BY MODERATOR)