We have Openalab CDS 2.4 (Update 06) and having some issues displaying uv purity for certain peaks. Some peaks are shown, while others are not. Is there a setting i am missing in processing method?
The first question how was the spectra collected? Did you collect all spectra, every 2nd, or use some type of peak detection to trigger collection. If you used peak detection can you manually check to see if the peak has collected spectra?
The spectra was collected for all from 200-400 nm
You might want to open a support ticket using your normal support channels. We would need to look at the data where you are seeing the issue and you likely do not want to post that here in the community.
I found the same symptoms. We're using Openalab CDS 2.4, and we don't have purity in certain peaks.Has the problem been solved by any chance?
I contacted the Agilent Laboratory Information Support, but I was asked to approach the hardware...
If you have solved it, please give me feedback. Please.
Retrieving data ...