I need to connect CAD detector to 1290 with Chromeleon software
Depends what type of modules you have. Normally with Infinity II modules, you will use an ERI cable for the connection. With Infinity I modules, you will use a remote cable. Normally, you will attach the cable to the autosampler as the injection valve switch will indicate the start of the run. Take a look at the back of your modules to see which cables would work in this case. I have attached the manual for the multisampler. See chapter 11 on page 319 for a list of cables and their part numbers.
Retrieving data ...