Our lab is using Agilent sea spray nebulizer for Agilent 5110. During our routine analysis for high salt samples, we have encountered sudden surge of the neblizer pressure and dropping of internal standard intensity so many time. I guess this could cause by clogging in the neblizer, as the problem can be solved by switching to the new one.
Does anyone know any durable nebulizer which can be utilised for Agilent 5110 against high salt solution? The sample is hydrochlic acid matriz with high contents of sodium and iron. Any other suggestion would be welcomed.