Followed the LAN setup in the manual, yet all I get is Powerfail. Using Chemstation version B.05.04 (which is supposed to work on Windows 10).
Hello monty_python66 :
Please see the attached communication troubleshooting guide. One possibility is that you do not have the addresses set to use the default LAN card address. The default address of the G1369x LAN card is 192.168.254.11 with switches 5&6 up. The switches are located on the card. Set your computer to 192.168.254.10.
The unit connects using a JetDIrect card - there are no dip switches on it. I've also tried connecting using the GPIB interface with the same result. I did manage to get it to work once by connecting to it using the old Win7 computer, then disconnecting it and connecting to the Win10 one. However, once rebooted the connection was lost. I've even installed Bootp but that hasn't helped.
The HP Jet Direct card was not supported on Agilent instruments after Windows XP.
Retrieving data ...