Can i use valve AVS with inert cyclonic camera in ICP-OES?
I assume you mean inert spraychamber? Yes, the AVS can be used with the inert spraychamber. Just a note on the inert spraychamber, the stabilisation time may need to be altered when using this as it can be a little different to the quartz cyclonic spraychamber.
The best way i have found to determine stabilisation time is to set it to 0, and run 60 reps at 1 second read time. You can then analyse a sample with elements of interest for you, and use the replicate intensities to determine at what point you signal stabilises. I test a few elements, as the stabilisation time can be slightly different for different elements.
Hope this helps.
Retrieving data ...