i have encountered an error with Cary 100 UV-VIS, E9328 fatal initialisation, fault reported by the instrument software while calibrating, error code:9:9328SEQ:E9328 fatal initialisation, this error occurs when initialising wavelength
The first step is to replace and align the lamps. If the problem persists then it's likely to be a micrometer issue. Please contact you local Agilent office for onsite service options.
Retrieving data ...