I have problems with the flow and pressure of my gas chromatograph.
Working with a gas chromatography equipment (Agilent 6890N) coupled to a mass spectrometer (Agilent 5973) and both the flow and the pressure fluctuate and do not stabilize, after doing an internal leak test of the injector, it gives us positive, we have changed everything and still the problem persists. The consumables we have changed are:
-Cu tubing from Splitless
-Liner Splitless: Gold Plated Inlet Seal with Washer, injector line septum.
-DB-5 GC column was cut in both extremes, inlet and MS line zones, in case there was any obstruction. At the same time, fittings and correspondent graphite ferrules were replaced.
-Split Vent Trap
Do you know where the problem may come from?
I know that I can use Argon gas as an indicator of leaks, but I would not know exactly how to do it, if you could also indicate that it would be of great help