Recently to accordance with my compliance i must to run IQOQ test in my Cary 630, but signal to noise has failed. The configuration is Transmission Module with ZnSe Optics, Firmware: 1,840, DLL Version: 1,431 and Software Version: 184.108.40.206.
The result to signal to noise test was 26546 and the software show me 30000 criteria, analyzing the user manual with this system configuration the criteria is 25000.
I did something wrong ? I tried to Agilent assistance but the technician told me he never saw that error.