first Turn off all gas flows at their sources and reconfiguration the Front inlet
Go to keyboard options to unlock
1 Press [Options]. 2 Select Keyboard and Display. 3 Select Hard Configuration lock, then press [No/Off].
on the GC keypad press config ,
Scroll to select
Front inlet than press Enter than press Mode/Type and removed the inlet module
after this please restart the GC system
and again fellow same procedure for reconfigure the front inlet and check
Thank you. I followed your directions and the problem has been solved.
we saw a similar problem (not negative septum purge flow) in that our purge flow could not get down to 3 ml/min. We could however get it to 4 ml/min.
We resolved it by replacing the EPC.