In EZChrom, you can define group (instead of named peak) in Peak or Group table. In Group, you need to choose "Calibrated Range" in Group type, you may define retention time range for individual peak, then keep Level 1 standard amount. Software will treat 3 peaks as whole to calculate RF.
Hope it helps.
Thank you very much, forwarded your suggestion to the customer, waiting for their feedback.