I recently had a problem with an hplc autosampler (Agilent 1100 series) in my lab. The indicator lamp shows red each time the instrument is turned on. What can be the possible causes and how can I fix it?
We are not allowed to give out service manuals. What model sampler is this? What does the error say?
I can give you part numbers. If it is the fan, then make sure it is in the same orientation when replaced.
You may want to call your local service group to repair.
Do you have anything hooked up to the remote port? What are you using to control your system? Check the logbook for errors.
Thanks Jhagel. The problem seems to be either a defective autosampler fan or electronics board. But how do I replace the autosampler fan or fix the defective electronic connection?
Retrieving data ...