I got through the hurdle!
You must choose the menu item "GC Configuration" and then select "get configuration from instrument".
The method that is currently loaded is updated and then saves over itself.
Then you can continue through the Wizard to set up the backflush.
Hopefully this is helpful to someone else.
Would you mind providing us with the revision of MassHunter Acquisition that you are running? You can determine this by going to Help>About.
I am using B.07.05 SR1.
Hello, I need help with an additional request regarding backflushing setup. Is there a way to enable constant backflushing during a method? My manager would like to enable backflushing when the instrument is not in use, so I want to create a "LAST.m" method to load at the end of a sequence that has backflushing enabled.
It appears that the inlet cannot have backwards flow at the beginning of a run, or it needs a flow between 3 and 1,250 mL/min when in split mode. I can switch to splitless, but I don't think that is backflushing out of the purge valve like we want it to, right?
Thanks for any insight.