How to generate calibration deviation report? any template please share.
Details: required to get the difference between the previous calibration and current calibration.. please assist to get this..
I marked this post as a question.
What revision of OpenLab CDS 2.x are you running? And would you mind elaborating more on the calibration comparison you want on the report?
Thank you for the reply,
I'm using OpenLab CDS 2.3,
when generating the calibration report it should show the previous calibration value & current value for a,b,c & required to compare the residual, purity of each compound in previous calibration & current calibration.
Since the calibration curve information is stored in each data file's results, you could set up this type of report by adding two of the Calibration Curve report items to a sequence summary report. You would then just need to set up filters on these groups so that they only display the calibration curve information for one particular injection (one group filtered to display the "old" curve information and the other group filtered to display the "new" curve information). These filters can be set up using a variety of expressions to limit results to specific Sample_CalibrationLevel, Sample_AcquisitionOrderNo, Sample_Name, etc...
Retrieving data ...