Hi KJNR, go to the dashboard tab and check the camera water flow and RF water flow making sure they are in the green zone. If one is in the green zone and the other in the red zone it suggests a blockage in the instrument. If both are low (or in the red zone) it could be you don't have enough flow from the chiller, especially if the chiller has long supply hoses. It maybe necessary to increase the water supply pressure to compensate.
To add on to Adrian. Make sure the chiller is full of water. If you still have the error, increase the pressure. The pressure can go up to 45 with no issue.
This is done by taking off the chiller's top cover(by removing two screws at the back of the chiller). There will be a visible handle you can turn that will adjust the pressure.
Thanks, this was resolved. Had to replace the manifold
Analytical Lab First Line Supervisor
ExxonMobil Fuels and Lubricants
1795 Burt Street, Lab #1106
Beaumont, Texas 77701
image001.jpg 332 bytes