To one and all,
I appear to have a bad power switch on my G1311A Quat pump. If I hold (or wedge) the button in, I have power. If I release it, I lose power. Has anyone experienced this and is there a reasonably easy fix without an Agilent repair crew... Sometimes my wedge pops out during runs and I lose power - and all connectivity my LC/MS.