Agilent 7100 CE with a firmware B.07.20 , continues to receive an error message Clean Level Sensor. Agilent Tech Service changed everything possible on the instrument.
the replenishment vial stops midway which thus creates an error. Did not have any problems before upgrade. Any Suggestions?
Normally this has to do with the level sensor and that has been replaced.
Could also be the firmware- you have B.07.20 and B.07.21 is the current one.
Does it happen to every method?
You may need to have the FSE go back to take a look.
Retrieving data ...