I used the Agilent 2100 Bioanalyzer to assess the quality of purified RNA. However, two error codes came up and prevent me from obtaining RINs.
I selected the prokaryotic total RNA assay.
The first code, of which I am more concerned, is code 4,501: "unexpected sample type". Does this mean my samples may be contaminated? There are some replicates which do not have this error, have an RIN of 10, and the electropherogram looks pretty much identical to the samples with the error code. Perhaps the electrodes have excess RNA zap on them that is contaminating the samples?
The second code is unexpected 5S subunit, which I've seen before and know how to bypass, but was wondering if the two codes may be related.
I would appreciate any insight!
Unfortunately, there is no Bioanalyzer section on the community at this point in time. Your best course of action at this point in time would be to contact the Agilent Bioanalyzer support team. You can reach them by emailing firstname.lastname@example.org .
NOTE: This support email address is for customers in North America. If you are from another region, then please use the following link to determine the appropriate Agilent contact information: Contact Us | Agilent