The sequence has run, but I get the error message 'no signals loaded' in Chemstation. Also, the log shows, CP Macros: Data not available:Data analysis not done. Can someone please help me understand this issue and possibly solve it?
What are your detector method parameters? Basically you haven't acquired any signal data. Do you see the red vertical line when your run starts? What version of software are you running?
Thank you for the reply. I did see the chromatograms in real time, during the run. However, the method was not set up to collect signal data (detector method parameters) as you said. So 'no signals loaded'.
What you might also consider doing is going into Windows Explorer (a.k.a. File Explorer in Windows 10) and navigating to the location of the files for which you're seeing the error message you stated. The typical location for ChemStation C.01.07 and earlier is C:\Chem32\1\Data\[your subdirectory or data result set folder]. The data "files" are really folders in ChemStation and they all have the extension .D. If you go into one of these folders corresponding with a datafile that has the problem, do you see any files with the extension .CH? This would be your signal file and its absence would indicate the signal wasn't collected at all.
Do you still need help on this issue? If so, please come back and let us know so we can continue trying to assist. If an answer provided helped you fix the problem, please log back in to the Community and mark it as "Correct Answer."
Retrieving data ...