Waiting on delivery of a Cary 60. Need to have an external machine provide a trigger to start a scan for spectral data (typ. 400nm to 700nm, in transmission) and output to a file. Is there a way to do this (USB, rs-232, etc?)
There is a plug inside the sample compartment on the left hand wall, that fits a 3.5mm phone ( audio) jack. All you would nee to do is connect a switch to a 3.5mm plug. Once the switch is triggered the Cary will START the application analysis.
To answer the question about outputting a file, in the Report Tab of each method, there is the option of doing and Autoconvert to either a ASCII (csv) or a ACII ( csv) with log file.
This will automatically convert the file at the end of the run.
greif, I just wanted to follow-up to see if your question was answered by dgrintuch's replies yesterday. If so, please come back to the poast and click the "Correct Answer" button on the response that helped so it will make the solution more visible. If you still need help, just let us know and we would be happy to continue working with you.
I went ahead and flagged the "Correct Answer" on this post. If you still need help though, please let us know and we would be happy to continue working with you.
Retrieving data ...