I have been trying to calibrate beyond the m/z range of the standard Agilent tuning mix, which only goes out to approximately m/z 2700. I put in CsI, but I only see the Cs+ ion and none of the adducts. Any thoughts?
The TOF calibration curve is linear out to 4000 m/z so the default calibration using m/z 2721.8948 should provide satisfactory results. I've never tried to calibrate an Agilent TOF with CsI but I suspect that the clusters might not stay intact through the ion optics pathway. You might try to lower the fragmentor voltage to start; the default settings may be de-clustering the CsI clusters.
I went ahead and flagged the “Correct Answer” on this post. If you still need help though, please let us know and we would be happy to continue working with you.
Retrieving data ...