3 people found this helpful
Below are the release notes for the new driver. New drivers are released periodically to allow us to use new equipment without having to upgrade a complete software package each time. The new driver has some enhancements for the older systems, but is really targeted towards new products including the new Intuvo 9000 GC and a new method of pressure control on the 7820 GC. If this doesn't answer your question, please respond with the GC model and current SW you are running, and I can advise further.
New in this Release
Support for the new Agilent Intuvo 9000 GC
Support for 78xx series instruments
Support for 7820 EPR
Enhance the user experience with user interface updates
Support Retention Time Locking on OpenLAB 2
Supported Data Systems
OpenLAB CDS (2.1, 2.0)
OpenLAB CDS ChemStation Edition (C.01.07, C.01.06, C.01.05)1
GC/MS MassHunter Acquisition (B.07.05, B.07.04, B.07.03, B.07.02)
OpenLAB CDS EZChrom Edition (A.04.07, A.04.06, A.04.05)
EZChrom Elite (3.3.2 SP2)
Multi-Technique ChemStation (B.04.03 SP2)
Issue Id KPR No. Title
256054 338 EZChrom: PALxt module not available after install
263680 339 Not allowed to set proper GC column max temp because it is less than 150 degrees.
264365 340 FID signal subtraction setting changes spontaneously from "Nothing" to "Column Compensation" during run
264809 341 "Extend Run" in GC Driver did not work correctly
265467 342 Cannot configure a 7890B on an existing client server system
267397 343 Copy to Clipboard Function in Instrument Actuals Throws Exception
267774 344 The column flow calculator calculation is off by almost 3x's the flow rate for the system.
267879 345 Sample Overlap settings can no more be edited after driver update, but are still part of the method
268821 346 Error Unhandled Exception when changing columns and a Gerstel CIS4 in installed in the 7890.