I am a new GC/TQ user. I am using a GCMS 8890 with a 7010B TQ system. I recently changed the column on the GC and ran some samples, I am continuously getting an elevated baseline, to cross out contamination, I have baked the column twice. Also, my baseline towards the end elevates as can be seen in the image attached. The masses are also given underneath.