I have a new GCMS and am using H2 carrier gas. I am seeing unusually high background at around 80,000 counts and poor tailing on all peaks. I have baked out the column and I have started to condition the source. Anyone with experience using H2 have advice to decrease these two issues so it's usable? Method parameters to use? The method is for drugs/controlled substances. Thank you!